If you have two or more instruments on the bus, you can disconnect all instruments except one to determine its address. Click on Scan for Instruments. Well, what if you were able to see a piece of equipment before you purchased it? With the VISAIC utility, you can speed up application development by learning how to automate measurements with your instruments, uncover GPIB problems, and avoid headaches by identifying malfunctioning instruments. We’ll send you an email letting you know exactly when your piece of equipment will be available for viewing Name: Instruments must be powered on and connected to the GPIB controller to be recognized. Please note that any additional items included with this equipment such as accessories, manuals, cables, calibration data, software, etc.
|Date Added:||15 April 2010|
|File Size:||22.75 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
At-gpi indicates that you should check for loose or disconnected cables, power disruption, or a malfunctioning instrument. Find an unused expansion slot in your computer.
National Instruments At-gpib/tnt Plug and Play Ieee Card d | eBay
Not af-gpib a picture from the manufacturer’s website, but the actual piece of equipment you would receive. To avoid such damage in handling your board, touch the antistatic plastic package to a metal part of your computer chassis before removing the board from the package.
When VISAIC initially runs, it automatically finds all of the available resources in the system and lists the instrument descriptors for each resource under the appropriate resource type. Click on Scan for Instruments.
Well, what if you were able to see a piece of equipment before you purchased it? Please contact us to complete your request.
You also can perform both a write and a read automatically by pressing the “Query” button. All logos and trademarks are properties of their respective companies. Click a thumbnail to view larger image: It’s like having the store come to you!
The basic functions and concepts described above may seem simple; however, they can be invaluable in troubleshooting and getting your GPIB system up and running. Valid Email Required Item to Inspect: With the VISAIC utility, you can speed up application development by learning how to automate measurements with your instruments, uncover GPIB problems, and avoid headaches by identifying malfunctioning instruments.
InstraView Request Form To get started: Now from the viRead tab, we select a count ofthe length of our expected response, and then click Execute.
Manuals, Datasheets, Drivers, Links.
AT-GPIB/TNT NATIONAL INSTRUMENTS Computer Accessories
Install Instrument Drivers Next. You have just confirmed communication with your instrument. The best way to achieve this is by using instrument drivers.
It might be a tight fit, but do not force the board into place. Stock Quick Quote.
National Instruments At-gpib/tnt Plug and Play Ieee-488.2 Card 182885d-01
If there is a problem, you no longer see your device. Make sure that your computer is powered off.
It is important to choose an alias that you can immediately identify with the intended instrument. Configuration in Windows tng USB.
Please contact one of our Customer Support Specialists if you have any questions about what is included with this equipment or if you require any additional information. To communicate with your instrument at Address 4, follow these instructions:.
National Instruments Ni at Gpib/tnt IEEE 488.2 182785e 01 GPIB Interface Card
Electrostatic discharge can damage several components on your GPIB board. Once you have completed these steps, you need a fast way to progress from interactive mode to programming mode so you can immediately begin writing your tests without a tedious transition process. Please note that any additional items included with this equipment such as accessories, manuals, cables, calibration data, software, etc. Unfortunately, engineers sometimes forget to reconnect all the cables and power up all the instruments.